TO LOAN THIS DEVICE PLEASE GO TO THE LAB SHOP
To begin using the data logger, connect the mini-USB cable to your data logger and an available USB port on your PC.
Setup the data logger by clicking ‘Setup and start the USB data logger’ and follow onscreen instructions.
When the setup is complete, the data logger should be removed from the USB port to begin logging.
ICA31 Multimeter Current Clamp Adapter AC, 400A ac, 29mm With RS Calibration
Set the DMM at ACV function and proper range.
Connect the plug of transducer with DMM firmly.
Press the trigger to open transformer jaws and clamp one conductor only, making sure the jaw is firmly closed around the conductor and reading the result from the digital display of DMM.